HP 3325A SERVICE MANUAL PDF
View and Download HP A calibration manual online. Synthesizer/Function Synthesizer HP A Operating And Service Manual. Frequency ( pages) . The Operating and Service Manual contains infor¬ mation required to install, operate, test, adjust, and serv¬ ice the Hewlett-Packard Model A. Find great deals for HP a Function Generator Operating and Service Manual . Shop with confidence on eBay!.
|Published (Last):||19 February 2008|
|PDF File Size:||8.85 Mb|
|ePub File Size:||9.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
Instruments with Option do not have rear panel signal output. The program used for this test is written specifically for the -hp- Model A Calculator but may be adapted to other controllers.
Repeat steps d and g. TB 4 Enter This procedure is for instruments with the Option High Stability frequency Reference.
Disconnect the ac line cord before replacing the fuse. A device will stop requesting service upon being Serial Polled, or if it no longer needs service. Ground the input and set the trace to the center line.
HP a Function Generator Operating and Service Manual | eBay
The output signal for each function is centered about zero volts unless a dc offset has been programmed. Go to subroutine to determine the reason.
Change function to negative slope ramp and repeat Step c. Alternating current power line. See Figure A.
HP 3325A Calibration Manual
Don’t have an account? Then press the O or key momentarily to increase or decrease the value of that digit by 1. Make the following A keyboard selections: Connect A signal output to CRO and voltage divider via a 6 ft.
Basic Block Diagram of Control 8-G-l. Maximum DC Offset Box 69, Marysville, Washington Printed: Note that the part number for A3U16 in this earlier design was The oscilloscope display should show a sine wave four 3325w peak-to-peak, one cycle per division.
Expand the oscilloscope vertical display and adjust controls so that the overshoot can be measured.
Set A amplitude modulation off. At 18MHz, check for the 3: Frequency automatically changes to 10kHz. Trigger CRO on A sync output. Change the A frequency to kHz. Does not cause SRQ. The instrument can be rack mounted with or without mahual handle kit by use of the following items: The data lines carry coded messages which represent addresses, program data, measurements, and status bytes.
Offset and amplitude incompatible 6.
HP A CALIBRATION MANUAL Pdf Download.
Rise and Fall Time: The tilt stand raises the front of the instrument for easier viewing of the control panel. Mnaual it, continue serbice print. Somerville, NJ Intersil Inc. Dimension a string variable for each parameter you want to interrogate. Earlier versions of this instrument, however, differ in design and appearance from those currently being produced. If the program string were corrected to make all data valid, this printout would result from the above program.