FRESENIUS 2008K MANUAL PDF
We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||23 June 2004|
|PDF File Size:||12.42 Mb|
|ePub File Size:||6.6 Mb|
|Price:||Free* [*Free Regsitration Required]|
What has changed, not the machine! Connecting The Extracorporeal Blood Circuit The k home Hemodialysis Machine Cleaning The Exterior Surface Following disinfection, they should be rinsed thoroughly with treated water. Modifying the K operation mxnual accommodate pediatric patients is accomplished by touching the Pediatric button on the Touch Screen.
Fresenius Medical Care 2008K Operator’s Manual
Single Needle Dialysis optional Notes: Post Temperature Sensor Calibration K Dial Valve Failure 1 Electronic frezenius failure. See Transmembrane Pressure maintenance Page 94 The result will be more accurate if both tests are done at the same blood flow rate.
The following example illustrates this procedure. An fresenihs thigh cuff is also available. Table Of Contents Check RO-processed water for residual disinfectant and be sure the water is within acceptable limits before connecting the water supply to the K.
New Fresenius k semi & annual pm manual – Topic
Page 74 Figure 34 — Heparin Delivery Screen on page Additional Monitoring Alarm in case of power failure. Page Start key a second time and the pump segment will be released from the pump head.
I think odog is right. The heparin pump should be monitored to verify the intended infusion during treatment. Preparing The Extracorporeal Blood Circuit An audible alarm will sound every two minutes to alert the operator that blood is sensed while the TX Clock is paused. Sensor Calibration Procedures Surprised non of you “F-ers” ever volunteered to be the lean facilitator for your clinics since it was suppossed to be a grass roots effort.
Fresenius Medical Care 2008K Manuals
Dialysate should already be verified for absence of disinfectant, verification of prescription, conductivity, and pH should also be confirmed. Observe the system for fluid leaks and discontinue treatment if you are unable to correct any fluid leak quickly.
Front Panel Controls Periodically check the power cord for damage fraying, over-heating, cuts, scrapes, etc. The blue lines on the machine are guides for the venous bloodline from the dialyzer to the patient. Page 25 mm Hg and the TMP alarm limits are spread fully open for 30 seconds. General Warnings And Cautions Heater is off during pre-rinse.
Cleaning and Disinfection Daily cleaning, chemical, and heat disinfection procedures should be performed to maximize the efficiency and minimize bacterial levels within the system.
Monitor Calibration Procedures Connect water inlet according to the specifications for the machine. The dialysate delivery machine can be used with a variety of blood-tubing configurations. Page 91 The prescribed target single-pool value, ranging from 0.
Page Wait until blood pressure module completes resetting and retry blood pressure reading.
Monitoring The Treatment Replacing The 9-volt Battery The transmembrane pressure TMP is equal to the venous pressure minus the dialysate pressure measured in mm Hg. K Venous Pressure Alarm Low pressure detected in the mabual Blood 1 Check venous tubing for kinked line, clotting or clamps.
Test Equipment And Supplies Needed If frewenius own most of the market, who are they selling to?? Connecting The Extracorporeal Blood Circuit Be sure the end of the segment clears the bottom of the yoke.
The Prime Amount method is a machine option that is set in the Service mode, and limits the amount of saline used in the priming procedure to a preset volume. Page 31 Line from venous drip chamber is connected to a transducer protector port P and attached here to provide venous blood pressure readings. Table Of Contents Conventions Page From Drain line, intermittent collection using a 3 Liter PD drain manuak as a collection device with a Safe-Lock connector optional.
K High Flow Error Possible balancing chamber problem.
Confirm if to min.